- Research article
- Open Access
Pediatric supraglottic airway devices in clinical practice: A prospective observational study
BMC Anesthesiology volume 17, Article number: 119 (2017)
Supraglottic airway devices (SGA) are commonly used in pediatric anesthesia and serve as primary or back-up devices for difficult airway management. Most SGA are marketed without proper clinical evaluation. The purpose of this study was to evaluate the performance of the pediatric LMA Supreme™, Air-Q® and Ambu® Aura-i™.
This prospective observational study was performed at Bern University Hospital, Switzerland. With ethics committee approval and a waiver for written informed consent 240 children undergoing elective surgery with an ASA class I-III and a weight of 5-30 kg were included. Three different pediatric supraglottic airway devices were assessed: The LMA Supreme™, Air-Q® and Ambu® Aura-i™. Primary outcome parameter was airway leak pressure. Secondary outcome parameters included first attempt and overall success rate, insertion time, fiberoptic view through the SGA, and adverse events. The primary hypothesis was that the mean airway leak pressure of each tested SGA was 20 cmH2O ± 10%.
None of the SGA showed a mean airway leak pressure of 20 cmH2O ± 10%, but mean airway leak pressures differed significantly between devices [LMA Supreme™ 18.0 (3.4) cmH2O, Air-Q® 15.9 (3.2) cmH2O, Ambu® Aura-i™ 17.3 (3.7) cmH2O, p < 0.001]. First attempt success rates (LMA Supreme™ 100%, Air-Q® 90%, Ambu® Aura-i™ 91%, p = 0.02) and overall success rates (LMA Supreme™ 100%, Air-Q® 91%, Ambu® Aura-i™ 95%, p = 0.02) also differed significantly. Insertion times ranged from 20 (7) seconds (Air-Q®) to 24 (6) seconds (LMA Supreme™, <p = 0.005). Insertion was rated easiest with the LMA Supreme™ (very easy in 97% vs. Air-Q® 70%, Ambu® Aura-i™ 72%, p < 0.001). Fiberoptic view was similar between the SGA. Adverse events were rare.
Airway leak pressures ranged from 16 to 18 cmH2O, enabling positive pressure ventilation with all successful SGA. The highest success rates were achieved by the LMA Supreme™, which was also rated easiest to insert.
Pediatric supraglottic airway devices (SGA) are being used with great success and over 50% of general anesthetic procedures are managed with SGA . A review article showed that compared to tracheal intubation, the use of SGA results in a decreased number of postoperative airway complications like desaturation, laryngospasm, coughing or breath holding . Also, a reduction in postoperative nausea and vomiting [3, 4] and faster recovery  has been reported. More advanced second generation pediatric SGA have been released, but often without evaluation in comparative, industry-independent studies prior to marketing. This is reflected by a survey of the Association of Paediatric Anaesthetists of Great Britain and Ireland in which 77% stated that trials assessing pediatric SGA were necessary . Interestingly, 88% preferred first generation pediatric SGA over second generation SGA . Clinical trials on pediatric SGA are scarce [5,6,7]. Many devices have been tested in adults [8, 9], but those results cannot be extrapolated to children.
This prospective observational study evaluates the performance of three pediatric SGA in 240 children: The pediatric LMA Supreme™ (LMA-S™, LMA Company, LeRocher, Victoria, Mahe, Seychelles), AirQ® (Cookgas® LLC, St. Louis, USA) and Ambu® Aura-i™ (Aura-i™, Ambu A/S, Ballerup, Denmark). Based on our previous study  and published literature , pediatric SGA are expected to provide airway leak pressures of about 20cmH2O. Our primary hypothesis was that the mean airway leak pressure of each tested SGA was 20 cmH2O ± 10%.
This prospective observational study evaluated the clinical performance of three pediatric SGA. It was performed at the Department of Anaesthesiology and Pain Therapy at the Inselspital, Bern University Hospital, Switzerland. Ethical approval was given by the Internal Review Board of the Bern University Hospital. The internal review board issued a waiver of written informed consent. The study was registered through the international trials registry ClinicalTrials.gov under the identifier NCT01625858.
Participants, inclusion and exclusion criteria
Data of 240 patients were prospectively obtained. Patients were boys and girls, 0–17 years old, weighing 5-30 kg, with an ASA physical status I-III, and scheduled for elective surgery under general anesthesia with an SGA and a planned operation time < 4 h. Exclusion criteria were risk of aspiration, body mass index >35 kg m−2, cervical spine disease, congenital anatomical malformation, known difficult airway, upper respiratory tract symptoms within 10 days, preoperative sore throat or poor dentition.
Three pediatric SGA were compared: The LMA-S™ and the two intubation SGA Air-Q® and Aura-i™ (Fig. 1). The LMA-S™ comes with a gastric channel, allowing for a functional separation of the digestive and respiratory tract. The AirQ® and the Aura-i™ do not feature a gastric channel, but are SGA designed for management of difficult airways and for intubation via the SGA. The SGA were used according to manufacturer’s recommendations and the size was determined by patients’ weight. To avoid selection bias, the specific SGA investigated was not chosen by the anesthesiologist in charge, but, depending on availability on stock, by a study nurse who was not involved in the anesthetic management of the child.
Anesthesia followed the standard operating procedures of the Children’s Hospital Bern and was standardized to guarantee equal depth of anesthesia. After premedication and preoxygenation, anesthesia was induced by inhalation of nitrous oxide and sevoflurane with final end-tidal sevoflurane concentrations of 6%, or intravenously with propofol (4 mg kg−1) and fentanyl (1–3 μg kg−1) and/or alfentanil (20 μg kg−1). Adequate depth of anesthesia was verified by loss of eyelash reflex, symmetric small pupils, and lack of physical reaction to jaw-thrust . Bag mask ventilation was provided and the SGA was lubricated with water-based K-Y Lubricating Jelly (Johnson & Johnson Medical Limited, Gargrave, Skipton, United Kingdom). An assistant performed the jaw-thrust manoeuvre. The SGA was then inserted by an attending, a resident or by a certified anesthesia nurse, supervised by one of the senior pediatric anesthesiologists. A maximum of three insertion attempts was performed and cuff pressure was set to 60 cmH2O using a manometer (VBM GmbH, Sulz, Germany or Rüsch GmbH, Kernen, Germany). SGA were secured with an adhesive tape around the SGA spanning both cheeks. All patients were ventilated using pressure controlled ventilation. Following clinical standards, the SGA were usually removed in a deeply sedated, spontaneously breathing patient. The time of removal ultimately depended on the decision of the anesthesiologist.
All data were recorded by a member of the research team who was not involved in clinical care. Recorded demographic and anesthesia-related data included sex, age, weight, height, ASA class, type of anesthesia induction, and duration of anesthesia.
Primary outcome was defined as mean airway leak pressure [cmH2O] after placement of the SGA [10,11,12]: Cuff pressure was set to 60cmH2O. Gas flow was 3 l min−1 and the expiratory valve was closed at 30cmH2O. The airway pressure at which a steady state was reached was determined.
Secondary outcome parameters were first attempt and overall success rate. Failures and reasons for device failure were noted. Insertion time was measured from face mask removal to successful ventilation of the lungs [13, 14]. Subjective difficulty of handling was evaluated by the anesthesiologist on a scale from 1 to 5 (1: very easy, 2: easy, 3: difficult, 4: very difficult, 5: impossible), as done before [8, 15, 16]. Feasibility of gastric catheter insertion was noted.
When deemed clinically desirable by the attending anesthesiologist, a fiberscope was inserted through the airway port and in these cases, fiberoptic view was graded as full view of the glottis (1), partial view of the glottis (2), only epiglottic structures seen (3), or no glottic/epiglottic structures visible (4), as done before [17,18,19]. Epiglottic downfolding was noted.
Airway interventions after initial successful ventilation were noted as well as adverse events like aspiration, regurgitation, bronchospasm, obstruction, dental, tongue or lip trauma, and blood on the removed SGA. The day after surgery patients and/or parents were interviewed about side effects like sore throat, hoarseness, dysphagia, tongue numbness, and postoperative nausea and vomiting (all graded as no/mild/moderate/severe).
Based on earlier studies, we expected leak pressures for pediatric second generation supraglottic airway devices to be around 20cmH2O [5, 10]. Our primary hypothesis was that the mean airway leak pressure of each tested SGA was 20 cmH2O ± 10%. More precisely this means that the 95% confidence interval (CI) for mean airway leak pressure of each tested SGA is within 20 cmH2O ± 10% (18–22 cmH2O). Given these expectations, a sample size calculation with a two-sided alpha level of 0.05 and a power of 0.9 calculated 63 patients per cohort necessary. To compensate for drop outs or missing data, we planned to include 80 patients per cohort.
Binary data were analysed by Chi square or by Fisher’s exact test if more than 20% of expected values were below 5. Ordinal data were evaluated using Kruskal-Wallis test. Normal distribution for continuous data was tested using Q-Q plots and Shapiro-Wilk W test. Independent samples Kruskal-Wallis test was used for analysis of non-parametric data and ANOVA for analysis of parametric data. For multiple comparisons of statistically significant data, Student’s t-test was used for parametric data and Mann–Whitney u-test for non-parametric data. Bonferroni correction was applied. We performed a subgroup analysis with subgroups according to children’s weight (subgroup 1: 5–9.9 kg; subgroup 2: 10–19.9 kg; subgroup 3: 20-30 kg).
Data are presented as numbers (%) for binary and ordinal data, as mean (SD) for parametric data, or as median (IQR) for nonparametric data. A probability of p < 0.05 was considered statistically significant. Data were analysed using Stata V.13.1 (StataCorp, College Station, TX, USA).
Data of 240 SGA insertions were evaluated. 201 children (84%) were male and 39 (16%) female. Distribution of gender and other demographic data did not differ between groups (p > 0.05, Table 1). Age ranged from 0.5 to 12.9 years. Weight ranged from 5.6 to 30.0 kg.
For none of the SGA did the 95% confidence interval of airway leak pressure range between 18 and 22 cmH2O, leading to rejection of the hypothesis for all three SGA. Mean airway leak pressures differed significantly between groups (p < 0.001) and were 18.0 (3.4) cmH2O with the LMA-S™, 15.9 (3.2) cmH2O with the Air-Q® and 17.3 (3.7) cmH2O with the Aura-i™ (Table 2). In the posthoc analysis, leak pressure with the Air-Q® was significantly lower than with the LMA-S™ or the Aura-i™.
First attempt and overall success rates differed significantly between groups (p = 0.02, Table 2). The LMA-S™ achieved a 100% first attempt success rate. This was statistically significantly better than the first attempt success rate of the Air-Q® or the Aura-i™. The overall success rate of the LMA-S™ was also statistically significantly better than that of the Air-Q®. For the LMA-S™, the 95% confidence interval of first attempt success rate was above 90% (Table 2).
Reasons for device failures were massive air leaks (7 Air-Q®; 4 Aura-i™), but no problems with insertion of the SGA. There was no difference in the number of insertion attempts (p = 0.28, Table 2).
Insertion time differed significantly between groups (p = 0.005, Table 2). The Air-Q® was fastest to insert and this was statistically significantly different to the LMA-S™.
Ease of insertion was best with the LMA-S™ (Table 2) and this was statistically significantly different to the other two SGA.
Gastric catheterisation was possible in 99% of LMA-S™ cases. Due to the lack of a gastric tube this was not possible with the Air-Q® or the Aura-i™. Fiberoptic view was similar with all three SGA (p = 0.41, Table 2). A full view of the glottis was achieved in 40% to 58%, and epiglottic downfolding was rare (10–17%, Table 2).
Several SGA leaked air or dislodged after initially successful ventilation, during lateral positioning for a caudal block (0 LMA-S™, 2 Air-Q®, 2 Aura-i™, p = 0.40) or intraoperatively (3 LMA-S™, 4 Air-Q®, 2 Aura-i™, p = 0.70). Of the leaks occurring intraoperatively, all leaks with the Air-Q® could be corrected by minor airway interventions like pushing or taping down intraoperatively, while all leaks with the LMA-S™ and the Aura-i™ led to removal of the SGA.
Adverse events and side effects
Two children regurgitated at emergence from anesthesia (Air-Q®). One child developed a laryngospasm requiring intubation and tracheal suctioning revealed some brownish secretion (Aura-i™). In all three cases, the postoperative course was absolutely uneventful with normal auscultation, continuous SpO2 > 95% at room air, and timely discharge. Several children (1 LMA-S™, 1 Air-Q®, 1 Aura-i™) developed a short period of slight obstruction during or after surgery. Laryngospasm developed with one Air-Q® and one Aura-i™ during surgery and with one LMA-S™ and two Aura-i™ after surgery. One child with an Aura-i™ with bronchospasm was intubated. All other spasms resolved spontaneously or by deepening the anesthesia. One child with an Air-Q® showed minor mucosal trauma. Blood stains were observed in 2 LMA-S™, 1 Air-Q® and 5 Aura-i™.
There was no difference in sore throat (3–4%, p = 0.76), hoarseness (3–9%, p = 0.37), numbness of the tongue (0–1%, p = 1.00), nausea (9–22%, p = 0.051), or vomiting (7–15%, p = 0.23) between groups. The only difference between groups was dysphagia (p = 0.01), resulting from a higher rate of dysphagia with the LMA-S™ (6%) versus the other groups (0%). This missed statistical significance in the posthoc comparison with Bonferroni correction.
Subgroup analysis was performed for subgroups according to weight (Table 3), as done previously . Since numbers in subgroup 1 (5–9.9 kg) were low statistical analysis was performed for subgroups 2 and 3 only. Fiberoptic view significantly differed between subgroups for the LMA-S™ and for the Aura-i™, with better views in children weighing 20-30 kg compared to children weighing 10–19.9 kg. Insertion times differed for the Aura-i™ with faster insertion in children weighing 10–19.9 kg compared to children weighing 20-30 kg (Table 3).
Comparison of the three SGA showed statistically significant differences in insertion time within subgroups 2 and 3, and for leak pressure within subgroup 3 (Table 3).
Confidence intervals of mean airway leak pressure were below the range of 18–22 cmH2O with all three SGA. Leak pressures ranged from 16 to 18 cmH2O and differed between devices. The Air-Q® showed the lowest leak pressures, which were similar to previous studies [6, 7, 20]. A previous study on the Ambu® Aura-i™ reported mean airway leak pressures of 16 cmH2O , while this was 18 cmH2O in our study. Leak pressure of the LMA Supreme™ was similar to a previous study reporting leak pressures of 20 cmH2O . Overall, leak pressures were lower than stated in our hypothesis, but still enabled positive pressure ventilation with all successful SGA and it is questionable whether the detected differences in leak pressure are of clinical relevance. Of note, our hypothesis was not based on a level of leak pressure that is associated with successful positive pressure ventilation of children, but on earlier results from other SGA [5, 10]. Taking the different studies into account, it seems that the range of measured leak pressures with different SGA is wide and starts at 16cmH2O. Leak pressures of SGA in children are therefore lower than those in adults [8, 21], but the clinically relevant point is that even with these low leak pressures positive pressure ventilation of children is possible.
We previously published a prospective study on the performance of the pediatric i-gel® and AuraOnce™ . In that study the leak pressure of the i-gel was 22 (5) and leak pressure of the AuraOnce™ was 19 (3) cmH2O. The study used the same methods regarding anesthesia and procedures as the current study, but included children weighing 5-50 kg. Other studies of the i-gel® showed similar leak pressures [22, 23]. Interestingly, leak pressure therefore seems to be highest with the i-gel®, which does not feature an inflatable cuff.
First attempt success rates in our study were lower than in other publications, which showed rates of 100% for the Aura-i™  and the Air-Q® [6, 7], and 96–97% for the LMA-S™ [5, 23]. Since SGA performance depends on children’s age and weight , these results might be explained by differences in demographics. In our previous study, first attempt success rates were 89% with the i-gel® and 90% with the Ambu® AuraOnce™, similar to data from the Air-Q® (90%) and Ambu® Aura-i™ (91%) in the present study. Thus, the LMA-S™ with its 100% first attempt success rate performed particularly well. Similarly, the ease of insertion, as graded by the person inserting the SGA on a commonly used subjective scale [9, 17, 18], was best with the LMA-S™.
We previously defined a first attempt success rate with a 95% confidence interval above 90% as the minimal desirable target for airway management devices . To our knowledge, no other definition of a minimal threshold for airway management devices exists. Several studies have shown success rates of ≥95% [5,6,7, 25], and we therefore postulated that a 95% confidence interval above 90% should be a realistic, achievable goal for airway management devices. In the present study, only the LMA-S™ achieved this target (95% CI of first attempt success rate 95–100%).
Fiberoptic view was similar between the three SGA in the present study, but a full view of the vocal cords was achieved in only 40–58%. In contrast, our previous study showed much better fiberoptic views with the i-gel® and the AuraOnce™ (full view in 88 and 89%, respectively) . The full view of the glottic opening from the orifice of the SGA is likely to reflect alignment of the SGA and the larynx and could play an important role for intubation through SGA. Interestingly, the Air-Q® and Aura-i™, but not the i-gel® and AuraOnce™, are specifically marketed for intubation. Success rates of 95–100% have been described for fiberoptically guided intubation through the Air-Q® [7, 26]. However, we showed that without fiberoptic guidance, intubation success rates through the pediatric Air-Q® and Aura-i™ are low, which is in line with the results of the fiberoptic assessment .
Air-Q® insertion was fastest in the present study. However, the faster insertion will most likely be of minimal clinical relevance.
In three children, anesthesia personnel reported regurgitation or possible aspiration. Clinical evaluation of minor aspiration remains difficult and it is unclear whether these children aspirated or not. In any case, their clinical course was completely uneventful. Temporary airway obstruction occurred in several children and rates were similar to reported rates . Most airway obstructions occurred towards the end of anesthesia. Due to the observational character of the study, the time of removal of the SGA varied and it is unclear how the depth of anesthesia influenced the occurrence of airway obstructions. Other adverse events like trauma or side effects such as sore throat were minor and rare.
More boys than girls were included in this study. This was due to a high percentage of circumcisions and likely does not influence outcomes. The ratio of boys and girls did not differ between groups.
The study was not randomized but observational, carrying the risk of selection bias. To prevent that anesthesiologists could choose a device of their liking, a study nurse who was not involved in anesthetic management of the patient chose the device. Also, demographic data were equally distributed between groups, showing that selection bias is rather unlikely.
Finally, our study included few patients weighing less than 10 kg (n = 8). SGA tend to perform differently in smaller and larger children [7, 10] and our results have to be interpreted with caution when transferred to younger patients. However, there was no statistical difference in age between the different SGA groups.
Leak pressure was lowest with the Air-Q®, but with leak pressures ranging from 16 to 18cmH2O, positive pressure ventilation was possible with all successfully inserted SGA. The LMA-S™ reached the highest first attempt and overall success rates, and was particularly easy to insert.
American Society of Anesthesiologists
Laryngeal Mask Airway Supreme
Supraglottic airway device
Bradley AE, White MC, Engelhardt T, Bayley G, Beringer RM. Current UK practice of pediatric supraglottic airway devices - a survey of members of the Association of Paediatric Anaesthetists of Great Britain and Ireland. Paediatr Anaesth. 2013;23:1006–9.
Luce V, Harkouk H, Brasher C, Michelet D, Hilly J, Maesani M, et al. Supraglottic airway devices vs tracheal intubation in children: a quantitative meta-analysis of respiratory complications. Paediatr Anaesth. 2014;24:1088–98.
Acquaviva MA, Horn ND, Gupta SK. Endotracheal Intubation Versus Laryngeal Mask Airway for Esophagogastroduodenoscopy in Children. J Pediatr Gastroenterol Nutr. 2014;59:54–6.
Zhao N, Deng F, Yu C. Anesthesia for pediatric day-case dental surgery: a study comparing the classic laryngeal mask airway with nasal trachea intubation. J Craniofac Surg. 2014;25:e245–8.
Jagannathan N, Sohn LE, Sawardekar A, Chang E, Langen KE, Anderson K. A randomised trial comparing the laryngeal mask airway Supreme with the laryngeal mask airway Unique in children. Anaesthesia. 2012;67:139–44.
Kim MS, Lee JH, Han SW, Im YJ, Kang HJ, Lee JR. A randomized comparison of the i-gel with the self-pressurized air-Q intubating laryngeal airway in children. Paediatr Anaesth. 2015;25:405–12.
Jagannathan N, Sohn LE, Sawardekar A, Gordon J, Shah RD, Mukherji II, et al. A randomized trial comparing the Ambu (R) Aura-i with the air-Q intubating laryngeal airway as conduits for tracheal intubation in children. Paediatr Anaesth. 2012;22:1197–204.
Theiler LG, Kleine-Brueggeney M, Kaiser D, Urwyler N, Luyet C, Vogt A, et al. Crossover Comparison of the Laryngeal Mask Supreme and the i-gel in Simulated Difficult Airway Scenario in Anesthetized Patients. Anesthesiology. 2009;111:55–62.
Sudhir G, Redfern D, Hall JE, Wilkes AR, Cann C. A comparison of the disposable Ambu AuraOnce Laryngeal Mask with the reusable LMA Classic laryngeal mask airway. Anaesthesia. 2007;62:719–22.
Theiler LG, Kleine-Brueggeney M, Luepold B, Stucki F, Seiler S, Urwyler N, et al. Performance of the pediatric-sized i-gel compared with the Ambu AuraOnce laryngeal mask in anesthetized and ventilated children. Anesthesiology. 2011;115:102–10.
Keller C, Brimacombe JR, Keller K, Morris R. Comparison of four methods for assessing airway sealing pressure with the laryngeal mask airway in adult patients. Br J Anaesth. 1999;82:286–7.
Kelly F, Sale S, Bayley G, Cook T, Stoddart P, White M. A cohort evaluation of the pediatric ProSeal laryngeal mask airway in 100 children. Paediatr Anaesth. 2008;18:947–51.
Lopez-Gil M, Brimacombe J, Garcia G. A randomized non-crossover study comparing the ProSeal and Classic laryngeal mask airway in anaesthetized children. Br J Anaesth. 2005;95:827–30.
Brimacombe J, Keller C, Brimacombe L. A comparison of the laryngeal mask airway ProSeal and the laryngeal tube airway in paralyzed anesthetized adult patients undergoing pressure-controlled ventilation. Anesth Analg. 2002;95:770–6.
Francksen H, Bein B, Cavus E, Renner J, Scholz J, Steinfath M, et al. Comparison of LMA Unique, Ambu laryngeal mask and Soft Seal laryngeal mask during routine surgical procedures. Eur J Anaesthesiol. 2007;24:134–40.
Theiler L, Gutzmann M, Kleine-Brueggeney M, Urwyler N, Kaempfen B, Greif R. i-gel supraglottic airway in clinical practice: a prospective observational multicentre study. Br J Anaesth. 2012;109:990–5.
Cook TM, Cranshaw J. Randomized crossover comparison of ProSeal Laryngeal Mask Airway with Laryngeal Tube Sonda during anaesthesia with controlled ventilation. Br J Anaesth. 2005;95:261–6.
Kapila A, Addy EV, Verghese C, Brain AI. The intubating laryngeal mask airway: an initial assessment of performance. Br J Anaesth. 1997;79:710–3.
Verghese C, Berlet J, Kapila A, Pollard R. Clinical assessment of the single use laryngeal mask airway--the LMA-unique. Br J Anaesth. 1998;80:677–9.
Jagannathan N, Sohn LE, Mankoo R, Langen KE, Roth AG, Hall SC. Prospective evaluation of the self-pressurized air-Q intubating laryngeal airway in children. Paediatr Anaesth. 2011;21:673–80.
Richez B, Saltel L, Banchereau F, Torrielli R, Cros AM. A new single use supraglottic airway device with a noninflatable cuff and an esophageal vent: an observational study of the i-gel. Anesth Analg. 2008;106:1137–9.
Jagannathan N, Sommers K, Sohn LE, Sawardekar A, Shah RD, Mukherji II, et al. A randomized equivalence trial comparing the i-gel and laryngeal mask airway Supreme in children. Paediatr Anaesth. 2013;23:127–33.
Kus A, Gok CN, Hosten T, Gurkan Y, Solak M, Toker K. The LMA-Supreme versus the I-gel in simulated difficult airway in children: a randomised study. Eur J Anaesthesiol. 2014;31:280–4.
Theiler L, Hermann K, Schoettker P, Savoldelli G, Urwyler N, Kleine-Brueggeney M, et al. SWIVIT--Swiss video-intubation trial evaluating video-laryngoscopes in a simulated difficult airway scenario: study protocol for a multicenter prospective randomized controlled trial in Switzerland. Trials. 2013;14:94.
Kim MS, Oh JT, Min JY, Lee KH, Lee JR. A randomised comparison of the i-gel and the Laryngeal Mask Airway Classic in infants. Anaesthesia. 2014;69:362–7.
Sinha R, Chandralekha RBR. Evaluation of air-Q intubating laryngeal airway as a conduit for tracheal intubation in infants--a pilot study. Paediatr Anaesth. 2012;22:156–60.
Kleine-Brueggeney M, Nicolet A, Nabecker S, Seiler S, Stucki F, Greif R, et al. Blind intubation of anaesthetised children with supraglottic airway devices AmbuAura-i and Air-Q cannot be recommended: A randomised controlled trial. Eur J Anaesthesiol. 2015;32:631–9.
The authors would like to thank Christine Riggenbach, RN, Franziska Stucki, MD, Stefan Seiler, MD, Yves Charriere, MD, and Marco Enderlin, MD (all Department of Anaesthesiology and Pain Therapy, Inselspital, Bern University Hospital, Switzerland) for their help with the clinical part of the study and data acquisition. Thanks to Ronald Martinez, BA, for English editing.
This work was funded by an institutional research grant of the Department of Anaesthesiology and Pain Therapy, Inselspital, Bern University Hospital, University of Bern, Switzerland.
Availability of data and materials
The datasets used and/or analysed during the current study are available from the corresponding author on reasonable request.
Ethics approval and consent to participate
Ethical approval for this study was provided by the Internal Review Board of the Bern University Hospital (Inselspital, Direktion Lehre und Forschung, 3010 Bern, Switzerland, Chairperson Prof. Kurt Laederach, approval number 12–011 May 15th 2012 and amendment approval number 13–040 March 28th 2013). The internal review board issued a waiver of written informed consent.
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Kleine-Brueggeney, M., Gottfried, A., Nabecker, S. et al. Pediatric supraglottic airway devices in clinical practice: A prospective observational study. BMC Anesthesiol 17, 119 (2017). https://doi.org/10.1186/s12871-017-0403-6
- Airway management
- Laryngeal mask airway
- Supraglottic airway device
- Pediatric anesthesia
- General anesthesia