- Research article
- Open Access
- Open Peer Review
The Alaris auditory evoked potential monitor as an indicator of seizure inducibility and duration during electroconvulsive therapy: an observational study
- Hsing-Hao Huang†1,
- Chun-Yu Wu†1,
- Feng-Sheng Lin1,
- Yi-Ping Wang1,
- Wei-Zen Sun1,
- Chih-Peng Lin1, 2Email author and
- Shou-Zen Fan1
© Huang et al.; licensee BioMed Central Ltd. 2014
- Received: 2 November 2013
- Accepted: 29 April 2014
- Published: 13 May 2014
Precise control of anesthetic depth during electroconvulsive therapy (ECT) is crucial because most intravenous anesthetics have anticonvulsant effects. In this study, we investigated the association between anesthetic depth measured by the Alaris auditory evoked potential index (AAI) and seizure inducibility and seizure duration during ECT.
Sixty-four ECTs were evaluated in 12 consecutive patients. General anesthesia was performed with a thiopental-based method. The relationship between the pre-ictal AAI, seizure activity and seizure duration was analyzed, and a possible threshold pre-ictal AAI to induce a seizure duration of at least 25 seconds was calculated.
Forty-one of the 64 ECT stimuli successfully induced seizure activity that lasted longer than 25 seconds. Pre-ictal AAI was significantly correlated to seizure duration (r = 0.54, p < 0.001) and the threshold pre-ictal AAi value was calculated to be 26 (area under curve: 0.76, sensitivity: 70.3% and specificity: 73.9%, p < 0.001). ECT with a pre-ictal AAI ≧ 26 had a higher incidence of successful seizure activity ( p < 0.001) and a longer seizure duration (55 ± 35 v.s. 21 ± 27 seconds, p < 0.001).
Maintenance of a pre-ictal AAI value ≧ 26 was associated with an increased incidence of successful seizure activities and a longer seizure duration. This is the first report to investigate Alaris AEP monitoring during ECT.
- Auditory evoked potential
- Electroconvulsive therapy
- Seizure inducibility
Electroconvulsive therapy (ECT) has gained importance over the years as a treatment for major affective disorders. This is partly due to the use of general anesthesia to reduce ECT-associated physical and psychological trauma . An induced seizure is necessary during ECT, and a seizure duration more than 25 seconds is often required for successful treatment . However, most intravenous anesthetics have negative effects on seizure duration due to their anticonvulsant effect. To optimize the potency of the ECT, a balance between adequate anesthesia and optimal duration of induced seizure is crucial. However, there is still a lack of consensus for a targeted level of anesthesia to maintain during the ECT.
Modern electroencephalogram (EEG)-based monitors such as bispectral index (BIS) and auditory evoked potential (AEP) allow a numeric description of the hypnotic component of anesthesia that by its instant availability enables the anesthesiologist to control anesthetic depth. Several reports have indicated a potential value of use of the BIS during the ECT [3–5]. In comparison, the potential role of AEP during ECT and the possible threshold AEP value needed to induce a successful seizure have not yet been clarified.
In the current study, we attempted to investigate the association between pre-ictal AEP score, incidence of seizure activity and seizure duration. In addition, we aimed to determine the possible threshold pre-ictal AEP score needed in order to induce successful seizure activity of more than 25 seconds.
This observational study was approved by the Research Ethics Committee of National Taiwan University Hospital. After explanation of the protocol, written informed consent was obtained from each patient. ECT was then prescribed to a consecutive 12 adult patients experiencing endogenous depression or schizophrenia. Each patient had not receive ECT before. Otherwise eligible patients who had with cardiovascular or cerebrovascular comorbidities or drug allergies were excluded. All patients were treated with ECT more than five times (three times per week at 2-day intervals).
The Alaris AEP monitor is a commercially available AEP monitor designed to estimate depth of anesthesia. It generates an "Alaris AEP index" (AAI), which is a dimensionless number scaled from 100 (awake) to 0. An AAI of about 60–100 represents awake status. Before induction of general anesthesia, the Alaris AEP monitor electrodes (AEP electrodes; Danmeter A/S), and a headphone for auditory stimuli were applied as recommended by the manufacturers. The three AEP electrodes were placed at mid forehead, left forehead (reference) and left mastoid. Pre-ictal AAI values were obtained in each ECT treatment and the data was analyzed later by an independent individual. General anesthesia was induced with thiopental (3–6 mg/kg), with the dosage and timing determined by an anesthesiologist unaware of the purpose of the study and blinded to the AAI score. Patient's airway is supported by facilitated face mask ventilation by attending anesthesiologist and a laryngeal mask is readily available. The electroshock stimulus was delivered by a trained psychiatrist using an ECT stimulator (Mecta spECTrum 5000Q, Mecta Corp, Lake Oswego, Ore). The electrical current was applied bilaterally for 1–4 seconds with a fixed pulse width (1.0 ms), a fixed pulse frequency (90 Hz) and a fixed pulse current (800 mAmp). A maximum number of shocks for each treatment is three times. Duration of each electroshock was adjusted in a stepwise method which began from 0.5 second. If there was no no seizure activity with the duration longer than 25 seconds and then the shock duration would be escalated to 0.75 second or 1.0 second for the second or third electroshock respectively. The efficacy of the electrical stimulation was determined by the so-called tourniquet technique—that is, the seizure activity was observed by convulsive movements of the distal leg, around which an inflated tourniquet was set to block the entry of the depolarizing muscle relaxant (succinylcholine, intravenous bolus 0.5-1 mg/kg) . The duration of each induced seizure was determined by monitoring the electroencephalogram seizure pattern and recorded. At the conclusion of the treatment, the patients were transferred to the recovery area after they had gained spontaneous breathing and could respond to verbal commands.
Those who performed data analysis was blinded to the treatment groups. The relationship between pre-ictal AAI and seizure duration was analyzed by the Pearson correlation coefficient test, and the threshold AAI needed o induce a successful seizure activity was determined by receiver operator characteristics curves (ROC), in which the area under the curve (AUC) was calculated. Wilcoxon rank-sum tests were applied to compare the pre-ictal AAI of successful or failed ECTs as well as the seizure durations of patients with a pre-ictal AAI above or below threshold value. The Fisher exact test was used to compare the incidence of successful seizure activity. The hypothesis was tested two-tailed at the p < 0.05 significance level with power > 0.9. All statistical analysis and graphs were performed using SigmaPlot for Windows, version 12 (SAS Institute, Cary, NC, USA).
Bipolar affective disorder
Body weight (Kg)
64.8 ± 9.2
165.8 ± 8.6
Seizure duration (second)
39.6 ± 36.0
Psychotropic medication (n, %)
Benzodiazepine type hypnotic
The average time for ECTs was 4.3 ± 2.0 minutes. The average thiopental dosage was 3.9 ± 0.94 mg/kg. Twenty-three of the 64 ECT stimuli failed to induce successful seizure activity, and the other 41 ECTs all induced a successful seizure activity longer than 25 seconds.
All patients were unable to respond to verbal command and lost the eyelash reflex within 1 min after injection of thiopental. No patient could recall the ECT procedures during the follow-up examination in the general ward.
The associations between pre-ictal AAI, seizure activity and seizure duration
The current study has shown that pre-ictal AAI is not only associated with seizure activity but also with seizure duration, and that a pre-ictal AAI of at least 26 may be a threshold for adequate anesthesia and successful seizure induction during the ECT. To our best knowledge, this is the first study to investigate the performance of the Alaris AEP monitor during ECT.
During ECT, it is not appropriate to merely administer intravenous anesthetics using a set dosage range, because the patients are usually receiving various chronic psychoactive drugs that may interfere with seizure inducibility . Because of this pre-existing variation in CNS depression, the intravenous anesthetic dosage may not be necessarily higher in the failed ECTs. This may be the reason that thiopental dosage was not significantly different between successful and failed ECTs in our study. Also, EEG-based monitors may have an important role as an overall gauge of CNS depression before ECT. Our study revealed a possible role of pre-ictal AAI in determining anesthetic depth appropriate for ECT, because the pre-ictal AAI was higher in the successful ECTs. Besides, by a high negative predictive property, 93% of shocks in patients with a pre-ictal AAI < 26 failed to induce a successful seizure activity. Therefore, a low pre-ictal AAI <26 should be avoided and so pre-ictal AAI may be a useful monitor to guide a more precise treatment. The efficacy of ECTs was previously believed to be related to a minimum seizure duration of 25 seconds and is currently believed to more related to occurrence of seizure . Among patients with a pre-ictal AAI less than 26, the incidence of successful seizure activity was lower and the mean seizure duration (about 21 seconds) was shorter than recommended. In comparison, among patients with a pre-ictal AAI above 26, there was a high incidence of successful seizure activities, and the mean seizure duration was far longer than the recommended length (about 55 seconds). The pre-ictal AAI above 26 was also within a safe range of that previously reported for unconsciousness which was reported about 35 in both adult and pediatric surgical patients [8, 9].
Both AAI and BIS can be considered effective measurements of anesthetic depth . Most studies evaluating anesthetic depth monitoring were performed in the context of longer procedures. In comparison, an episode of ECT has a duration not more than several minutes long, so more rapidly responsive monitoring is required. Because of the average time delay of BIS is more than 30 seconds , this delay may limit the usefulness of BIS during ECT. In contrast to BIS, the AAI monitor allows extraction of the AEP signal within 15 to 25 sweeps of 110-msec duration; a process that results in only a 6-second delay . Therefore, the AAI monitor may be an useful tool to determine a more precise timing for the electric shock of ECT. It has also reported that the BIS value that represents the awake state might become changed after the initial ECT . In comparison, the AEP may be more resistant to the electroshock wave [14, 15]. However, such characteristics of BIS and AEP during ECT need further investigation to confirm their validity.
There are several limitations in our study. First, because no previous report investigating AAI during ECTs, a convenience sampling is arranged in this observational study. The patient number is also small because that this population is relatively difficult to enroll. Therefore, the real value of usage of pre-ictal AAI during ECT may only be ensured by a randomized-controlled study, and our study may be a potential reference for further studies. Second, because methohexital is not available in our country, we used thiopental for ECT in our institute. Because different intravenous anesthetics have different proconvulsant and anticonvulsant effects , the threshold pre-ictal AAI value determined in this study may not be applicable to other intravenous anesthetics.
In conclusion, this is the first report to investigate Alaris AEP monitoring during ECT. To maintain a pre-ictal value at least 26 was associated with an increased incidence of successful seizure activity and a longer seizure duration.
- Ding Z, White PF: Anesthesia for electroconvulsive therapy. Anesth Analg. 2002, 94 (5): 1351-1364. 10.1097/00000539-200205000-00057.PubMedGoogle Scholar
- Rasmussen K: The practice of electroconvulsive therapy: recommendations for treatment, training, and privileging (second edition). J ECT. 2002, 18 (1): 58-59. 10.1097/00124509-200203000-00015.PubMedGoogle Scholar
- White PF, Rawal S, Recart A, Thornton L, Litle M, Stool L: Can the bispectral index be used to predict seizure time and awakening after electroconvulsive therapy?. Anesth Analg. 2003, 96 (6): 1636-1639. table of contentsPubMedGoogle Scholar
- Gunawardane PO, Murphy PA, Sleigh JW: Bispectral index monitoring during electroconvulsive therapy under propofol anaesthesia. Br J Anaesth. 2002, 88 (2): 184-187. 10.1093/bja/88.2.184.PubMedGoogle Scholar
- Hanss R, Bauer M, Bein B, Goeder R, Buttgereit B, Schulz-Du Bois AC, Steinfath M, Scholz J: Bispectral index-controlled anaesthesia for electroconvulsive therapy. Eur J Anaesthesiol. 2006, 23 (3): 202-207. 10.1017/S026502150500219X.PubMedGoogle Scholar
- Murali N, Saravanan ES, Ramesh VJ, Gangadhar BN, Jananakiramiah N, Kumar SS, Christopher R, Subbakrishna DK: An intrasubject comparison of two doses of succinylcholine in modified electroconvulsive therapy. Anesth Analg. 1999, 89 (5): 1301-1304. 10.1213/00000539-199911000-00043.PubMedGoogle Scholar
- Rasmussen KG, Laurila DR, Brady BM, Lewis CL, Niemeyer KD, Sun NM, Marienau ME, Hooten WM, Schroeder DR, Spackman TM: Seizure length with sevoflurane and thiopental for induction of general anesthesia in electroconvulsive therapy: a randomized double-blind trial. J ECT. 2006, 22 (4): 240-242. 10.1097/01.yct.0000244243.10167.b4.PubMedGoogle Scholar
- Gajraj RJ, Doi M, Mantzaridis H, Kenny GN: Comparison of bispectral EEG analysis and auditory evoked potentials for monitoring depth of anaesthesia during propofol anaesthesia. Br J Anaesth. 1999, 82 (5): 672-678. 10.1093/bja/82.5.672.PubMedGoogle Scholar
- Weber F, Bein T, Hobbhahn J, Taeger K: Evaluation of the Alaris Auditory Evoked Potential Index as an indicator of anesthetic depth in preschool children during induction of anesthesia with sevoflurane and remifentanil. Anesthesiology. 2004, 101 (2): 294-298. 10.1097/00000542-200408000-00008.PubMedGoogle Scholar
- Bruhn J, Myles PS, Sneyd R, Struys MM: Depth of anaesthesia monitoring: what's available, what's validated and what's next?. Br J Anaesth. 2006, 97 (1): 85-94. 10.1093/bja/ael120.PubMedGoogle Scholar
- Bonhomme V, Hans P: Monitoring depth of anaesthesia: is it worth the effort?. Eur J Anaesthesiol. 2004, 21 (6): 423-428. 10.1097/00003643-200406000-00002.PubMedGoogle Scholar
- Jensen EW, Nygaard M, Henneberg SW: On-line analysis of middle latency auditory evoked potentials (MLAEP) for monitoring depth of anaesthesia in laboratory rats. Med Eng Phys. 1998, 20 (10): 722-728.PubMedGoogle Scholar
- Thimmaiah R, Thirthalli J, Ramesh VJ, Radhakrishnan MC, Muralidharan K, Mahadevaiah M, Gangadhar BN, Umamaheshwara Rao GS: Effect of a course of electroconvulsive therapy on interictal bispectral index values: a prospective study. J ECT. 2012, 28 (1): 20-23. 10.1097/YCT.0b013e318235b658.PubMedGoogle Scholar
- Shaw NA: The effects of electroconvulsive shock on the primary cortical auditory potential. Biol Psychiatry. 1998, 43 (8): 595-600. 10.1016/S0006-3223(97)00446-0.PubMedGoogle Scholar
- Griskova I, Dapsys K, Andruskevicius S, Ruksenas O: Does electroconvulsive therapy (ECT) affect cognitive components of auditory evoked P300?. Acta Neurobiol Exp (Wars). 2005, 65 (1): 73-77.Google Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2253/14/34/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.