Tracheal tube and laryngeal mask cuff pressure during anaesthesia - mandatory monitoring is in need
© Rokamp et al; licensee BioMed Central Ltd. 2010
Received: 2 May 2010
Accepted: 3 December 2010
Published: 3 December 2010
To prevent endothelium and nerve lesions, tracheal tube and laryngeal mask cuff pressure is to be maintained at a low level and yet be high enough to secure air sealing.
In a prospective quality-control study, 201 patients undergoing surgery during anaesthesia (without the use of nitrous oxide) were included for determination of the cuff pressure of the tracheal tubes and laryngeal masks.
In the 119 patients provided with a tracheal tube, the median cuff pressure was 30 (range 8 - 100) cm H2O and the pressure exceeded 30 cm H2O (upper recommended level) for 54 patients. In the 82 patients provided with a laryngeal mask, the cuff pressure was 95 (10 - 121) cm H2O and above 60 cm H2O (upper recommended level) for 56 patients and in 34 of these patients, the pressure exceeded the upper cuff gauge limit (120 cm H2O). There was no association between cuff pressure and age, body mass index, type of surgery, or time from induction of anaesthesia to the time the cuff pressure was measured.
For maintenance of epithelia flow and nerve function and at the same time secure air sealing, this evaluation indicates that the cuff pressure needs to be checked as part of the procedures involved in induction of anaesthesia and eventually checked during surgery.
During general anaesthesia, pulmonary ventilation is secured with a tracheal tube or by a laryngeal mask and attention to the risk of complications related to a high intracuff pressure is important. When the cuff to tracheal wall pressure exceeds the tracheal capillary pressure (27-40 cm H2O) for approximately 15 min, the tracheal mucous membrane becomes ischemic . The intracuff pressure approximates the cuff to tracheal wall pressures in high volume/low pressure cuffs  and a cuff pressure below 30 cm H2O is recommended to prevent ischemic injury [1, 3]. Also recurrent laryngeal nerve palsy has been demonstrated in up to 5% of patients after intubation and a high cuff pressure is suspected to be important in that regard [4, 5]. Similarly in patients provided with a laryngeal mask, a high cuff pressure may lead to palsy of the lingual, hypoglossal, and recurrent laryngeal nerves [6–8] but with the cuff pressure maintained below 60 cm H2O, the airway seal is optimized [9, 10] and the incidence of a postoperative sore throat is low [11–13].
Here patients, provided with either a tracheal tube or a laryngeal mask during elective surgery requiring general anaesthesia, were assessed for the established cuff pressure. We aimed at evaluating the incidence of a cuff pressure that was outside the recommended level and hypothesized that especially overweight patients would be exposed to a high cuff pressure since a high body mass index caries a risk of gastroesophageal reflux  and that such patients require a high peak inspiratory pressure during mechanical ventilation .
The study was evaluated by The Scientific Ethics Committee of the Capital Region of Denmark (Journal no. H-4-2010-075) but not considered to require ethical approval since it was directed to quality control and did not involve any experimental procedures. We determined the cuff pressure in consecutetively enrolled patients from two hospitals: 97 from Herlev Hospital and 104 from Rigshospitalet. We included adult patients planned for operation in general anaesthesia and who were provided either with a tracheal tube or a laryngeal mask, while we excluded patients who had been intubated prior to arriving at the operating room and those for whom the airway was kept patent with a double-lumen tracheal tube.
When the patient arrived at the operating ward, the staff prepared the patient for anaesthesia and surgery according to local instructions and the anaesthesiologist in charge of the patient initiated the anaesthesia together with an anaesthesia nurse. For ventilatory and cardiovascular monitoring a Dräger CATO (type M32040, Lübeck, Germany) or Dräger Primus (G18155) anaesthesia machine was used together with a Phillips Intillivue MP70 monitor. Neuromuscular function was evaluated with a TOF monitor (Organon Dublin, Ireland). For induction of anaesthesia propofol or thiopental was administered guided by the weight of the patient and the administration was continued until the cilia reflex was eliminated. Anaesthesia was maintained with propofol, sevoflurane or desflurane and fentanyl or remifentanil was used for analgesia, while cisatracurium, rocuronium or suxamethonium facilitated tracheal intubation. The airway was kept patent with a tracheal tube (high volume/low pressure; Unomedical, Copenhagen, Denmark; n = 119) or a laryngeal mask (AuraOnce; Ambu A/S, Ballerup, Denmark; n = 82) for mamma (n = 49), gastrointestinal (n = 33), gynaecological (n = 11), orthopaedic (n = 5), plastic (n = 11), urological (n = 71), hepatic (n = 6), or vascular surgery (n = 15). Inflation of the cuff was not described in the local instructions for induction of anaesthesia and therefore carried out according to the disposition of the anaesthesiologist in charge of the patient and without the use of a manometer or a pressure release valve. In regard to this study there were no restrictions to the treatment of the patient before or after the cuff pressure was measured. According to local tradition, the airway cuff pressure is not adjusted during the surgical intervention and manipulation of the head does not regularly provoke a recheck of the cuff pressure.
When anaesthesia was established and the tracheal tube or the laryngeal mask was in place, the cuff pressure was determined by a Universal cuff pressure gauge with an upper scale limit of 120 cm H2O (VBM Medizintechnik GmbH, Baden-Württemberg, Germany). The manometer was connected to the pilot balloon and the cuff pressure was read from the manometer and documented together with the height, weight, age, and sex of the patient and the type of surgery and the time from placement of the tube or mask to when the cuff pressure was measured. In case the cuff pressure was outside the recommended level [3, 12], the pressure was adjusted.
Considering that we should detect 10 patients provided with a too high laryngeal or tracheal tube pressure in each department and the incidence is about 25% , we evaluated the cuff pressure in 200 patients. Data are presented as median and range and a p-value < 0.05 was considered to be statistical significant. For correlations between cuff pressure and age, body mass index, type of surgery, and time from induction of anaesthesia to determination of the cuff pressure, Pearson's correlation was used.
There was no significant relation between tracheal tube cuff pressure and age (r = 0.028), body mass index (r = 0.245), type of surgery (r = -0.001), or the time from induction of anaesthesia to determination of the cuff pressure (r = -0.168). Furthermore there was no significant relation between laryngeal mask cuff pressure and age (r = 0.129), body mass index (r = -0.015), type of surgery (r = -0.177), or the time from induction of anaesthesia to when the cuff pressure was determined (r = -0.074).
Discussion and Conclusion
In contrast to what we expected, overweight patients were not more frequently exposed to a high cuff pressure than other patients but even without the use of nitrous oxide for maintained anaesthesia, the cuff pressure exceeded the recommended level for about half of the patients provided with a tracheal tube and for almost three quarters of those patients provided with a laryngeal mask. We expected a high cuff pressure to be predominant in obese patients since a high body mass index caries a risk of gastroesophageal reflux  and a high peak inspiratory pressure during mechanical ventilation . A low laryngeal mask cuff pressure secures the airway seal [11, 10] and reduces the incidence of postoperative sore throat . It was therefore surprising to us that the majority of the patients were exposed to a too high cuff pressure and the most frequently measured laryngeal cuff pressure was in fact at a level that exceeded 120 cm H2O, arguing for that cuff pressure is an underestimated issue in general anaesthesia practice.
Sengupta et al.  found that among 93 patients provided with a tracheal tube and undergoing general anaesthesia, 50% of the patients were having a cuff pressure above 30 cm H2O and 27% had a cuff pressure above 40 cm H2O. Similarly in a prehospital setting Galinski et al.  found that among 107 patients, the tracheal tube cuff pressure was larger than 27 cm H2O in 79% of the patients. Thus with inflation of air into the cuff until air sealing, the anaesthesiologist has a poor ability to estimate a correct cuff pressure by palpation of the pilot balloon [18, 19]. When using inflation of a fixed volume of air into the cuff, a linear relationship between cuff volume and pressure is established , but the volume of air required (4.5 ml) to reach 50 cm H2O is only 50% larger than that required for establishing the safe tracheal tube cuff pressure of 30 cm H2O (3 ml), i.e. the safety margin is low.
Focus on the cuff pressure is often directed to patients exposed to nitrous oxide during anaesthesia. In these patients cuff pressure monitoring with automatic tracheal tube cuff pressure control is both reliable and stable  and for paediatric patients intubated with a cuffed tube, the use of a pressure release valve prevents that high tube cuff pressures develop [22, 23]. Use of a pressure release valve is relevant especially when a high compliance tracheal tube cuff is used, since it does not prevent a high cuff pressure to be transmitted to epithelia and the thin polyurethane membrane facilitates transmembrane diffusion of nitrous oxide with following rapid increase of the cuff pressure .
The window of application of a correct laryngeal mask cuff pressure is broader (< 60 cm H2O) and estimation of the cuff pressure by palpation of the pilot balloon is acceptable after some training . Still the use of a cuff pressure gauge is favoured to confirm that a correct cuff pressure is established, both in regard to a tracheal tube and a laryngeal mask, considering the low cost of the device (~100 €).
Among the limitations of this study is it that the cuff pressure was measured only once during anaesthesia and fluctuations in pressure by movement of the patient's head and change in the depth of anaesthesia or the level of neuromuscular blockade are not taken into account. It could be argued that the pressure reported is not representative for the whole period of anaesthesia, but measurements were spread over 358 min to obtain values both at induction of anaesthesia and during surgery. Also we did not register the technique used for inflation of air into the cuff, and we do not know whether the high incidence of cuff pressures above the recommended level is due to that the staff was unaware of the correct cuff pressure, or whether they did not inflate the cuff only to the level that secured the airway. To generalize the results of this study can also be argued: We only included two hospital departments, but a high incidence of cuff pressures above the recommended level is reported by others [16, 17, 26].
In conclusion this study demonstrates that more than 50% of patients are provided with a too high tracheal tube or laryngeal mask cuff pressure. We consider that to establish the correct cuff pressure is an important element in the anaesthetic procedures and that it involves the determination of the tracheal tube or laryngeal mask cuff pressure, not only at the induction of anaesthesia but also during its maintenance.
- Nordin U: The trachea and cuff-induced tracheal injury. An experimental study on causative factors and prevention. Acta Otolaryngol Suppl. 1977, 345: 1-71.PubMedGoogle Scholar
- Carrol R, Hedden M, Safar P: Intratracheal cuffs: performance characteristics. Anesthesiology. 1969, 31: 275-81.View ArticleGoogle Scholar
- Seegobin RD, van Hasselt Gl: Endotracheal cuff pressure and tracheal mucosal blood flow: endoscopic study of effects of four large volume cuffs. Br Med J. 1984, 288: 965-8.View ArticleGoogle Scholar
- Friedrich T, Hänsch U, Eichfeld U, Steinert M, Staemmler A, Schönfelder M: Recurrent laryngeal nerve paralysis as intubation injury?. Chirurg. 2000, 71: 539-44.View ArticlePubMedGoogle Scholar
- Wason R, Gupta P, Gogia AR: Bilateral adductor vocal cord paresis following endotracheal intubation for general anaesthesia. Anaesth Intensive Care. 2004, 32: 417-8.PubMedGoogle Scholar
- Endo K, Okabe Y, Maruyama Y, Tsukatani T, Furukawa M: Bilateral vocal cord paralysis caused by laryngeal mask airway. Am J Otolaryngol. 2007, 28: 126-9.View ArticlePubMedGoogle Scholar
- Stewart A, Lindsay WA: Bilateral hypoglossal nerve injury following the use of the laryngeal mask airway. Anaesthesia. 2002, 57: 264-5.View ArticlePubMedGoogle Scholar
- Brimacombe J, Clarke G, Keller C: Lingual nerve injury associated with the ProSeal laryngeal mask airway: a case report and review of the literature. Br J Anaesth. 2005, 95: 420-3.View ArticlePubMedGoogle Scholar
- Keller C, Puehringer F, Brimacombe J: The influence of cuff volume on oropharyngeal leak pressure and fibreoptic position with the laryngeal mask airway. Br J Anaesth. 1998, 81: 186-7.View ArticlePubMedGoogle Scholar
- Brimacombe J, Keller C, Morris R, Mecklem D: A comparison of the disposable versus the reusable laryngeal mask airway in paralyzed adult patients. Anesth Analg. 1998, 87: 921-4.PubMedGoogle Scholar
- Burgard G, Mollhoff T, Prien T: The effect of laryngeal mask cuff pressure on postoperative sore throat incidence. J Clin Anesth. 1996, 8: 198-201.View ArticlePubMedGoogle Scholar
- Brimacombe J, Holyoake L, Keller C, Brimacombe N, Scully M, Barry J, Talbutt P, Sartain J, McMahon P: Pharyngolaryngeal neck and jaw discomfort after anaesthesia with the face mask and laryngeal mask airway at high and low cuff volumes in males and females. Anesthesiology. 2000, 93: 26-31.View ArticlePubMedGoogle Scholar
- Wong JG, Heaney M, Chambers NA, Erb TO, von Ungern-Sternberg BS: Impact of laryngeal mask airway cuff pressures on the incidence of sore throat in children. Paediatr Anaesth. 2009, 19: 464-9.View ArticlePubMedGoogle Scholar
- Wajed SA, Streets CG, Bremner CG, DeMeester TR: Elevated body mass disrupts the barrier to gastroesophageal reflux. Arch Surg. 2001, 136: 1014-8.View ArticlePubMedGoogle Scholar
- Pelosi P, Croci M, Ravagnan I, Tredici S, Pedoto A, Lissoni A, Gattinoni L: The effects of body mass on lung volumes, respiratory mechanics, and gas exchange during general anaesthesia. Anesth Analg. 1998, 87: 654-60.PubMedGoogle Scholar
- Sengupta P, Sessler DI, Maglinger P, Wells S, Vogt A, Durrani J, Wadhwa A: Endotracheal tube cuff pressure in three hospitals, and the volume required to produce an appropriate cuff pressure. BMC Anesthesiol. 2004, 4: 8-View ArticlePubMedPubMed CentralGoogle Scholar
- Galinski M, Tréoux V, Garrigue B, Lapostolle F, Borron SW, Adnet F: Intracuff pressures of endotracheal tubes in the management of airway emergencies: the need for pressure monitoring. Ann Emerg Med. 2006, 47: 545-7.View ArticlePubMedGoogle Scholar
- Hoffman RJ, Parwani V, Hahn IH: Experienced emergency medicine physicians cannot safely inflate or estimate endotracheal tube cuff pressure using standard techniques. Am J Emerg Med. 2006, 24: 139-43.View ArticlePubMedGoogle Scholar
- Stewart SL, Secrest JA, Norwood BR, Zachary R: A comparison of endotracheal tube cuff pressures using estimation techniques and direct intracuff measurement. AANA J. 2003, 71: 443-7.PubMedGoogle Scholar
- Hoffman RJ, Dahlen JR, Lipovic D, Stürmann KM: Linear Correlation of Endotracheal Tube Cuff Pressure and Volume. West J Emerg Med. 2009, 10: 137-9.PubMedPubMed CentralGoogle Scholar
- Kunitz O, Jansen R, Ohnsorge E, Haaf-vonBelow S, Schulz-Stübner S, Rossaint R: Cuff pressure monitoring and regulation in adults. Anaesthesist. 2004, 53: 334-40.View ArticlePubMedGoogle Scholar
- Dullenkopf A, Bernet-Buettiker V, Maino P, Weiss M: Performance of a novel pressure release valve for cuff pressure control in pediatric tracheal tubes. Paediatr Anaesth. 2006, 16: 19-24.View ArticlePubMedGoogle Scholar
- Weiss M, Dullenkopf A, Fischer JE, Keller C, Gerber AC: Prospective randomized controlled multi-centre trial of cuffed or uncuffed endotracheal tubes in small children. Br J Anaesth. 2009, 103: 867-73.View ArticlePubMedGoogle Scholar
- Dullenkopf A, Gerber AC, Weiss M: Nitrous oxide diffusion into tracheal tube cuffs: comparison of five different tracheal tube cuffs. Acta Anaesthesiol Scand. 2004, 48: 1180-4.View ArticlePubMedGoogle Scholar
- Keller C, Brimacombe JR: Laryngeal mask airway intracuff pressure estimation by digital palpation of the pilot balloon: a comparison of reusable and disposable masks. Anaesthesia. 1999, 54: 183-186.View ArticlePubMedGoogle Scholar
- Braz JR, Navarro LH, Takata IH, Nascimento Júnior P: Endotracheal tube cuff pressure: need for precise measurement. Sao Paulo Med J. 1999, 117: 243-7.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2253/10/20/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (<url>http://creativecommons.org/licenses/by/2.0</url>), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.